The lubricating head adapter for NX adapter (EndoStationmini) is designed to enable effective lubrication (oiling) of both the RDT3-NX handpiece head (for the SAF System) and the rotary handpiece head (for Rotary instruments).
Oiling of any of the heads should be done through the micromotor adapter, and not independently. The shaft of the micromotor adapter should be inserted into the narrow orifice of the lubricating adapter, and the oil spray nozzle should be inserted into the wide orifice of the lubricating adapter. Oil should be sprayed for 1-2 seconds, until clear oil comes out of the head of the handpiece head (SAF or Rotary). If the oil comes out dark or unclear - repeat this procedure.
It is advised to use percaution while oiling, including the use of gloves, eye protection and a robe. The lubricating adater should be held using a dry cloth or paper towel.
Disinfection: Externally by alcohol wipes / disinfectant spray.
* Please note that the RDT3-NX handpiece head should be oiled after every patient, to avoid corrosion damage by the NaOCl irrigation.